Tagged: PEEP valve
- March 27, 2020 at 11:14 #98885
We are a university group of engineers working on a new version with the developers of this project: https://www.oxygen.protofy.xyz/
Due short stock and slow production of companies making the PEEP valves the respirator needs we need urgent help to design a PEEP valve we can easily manufacture with 3d printing or else.
Commercial peep valves looks like this
This article is very useful to understand the need and the way a PEEP valve works
EspañaMarch 27, 2020 at 12:11 #98888Filippo MenconiParticipant
Hello, I will start working on the valve in a few hours. I’m an aerospace engineer with a strong experience experience in mechanical design.
I will work on the 3D model of the valve. If somebody has started working on this device please feel free to write me at firstname.lastname@example.org, we could share the results.
FilippoMarch 27, 2020 at 12:44 #98894Ajay AggarwalParticipant
If anyone does get this design working, please share as we are working on developing the Decathlon Mask ventilator following the Italian design (Isinovva) but are falling short on PEEP valves
AjayMarch 28, 2020 at 17:35 #98973Sebastian EhingerParticipant
I’m also interested in the design if anybody could get this to work..
SebastianMarch 29, 2020 at 19:39 #98989Tom BrokampParticipant
You don’t need a 3D printer for this task. You need just
a) the hose, were the air should flow out
b) a glass, 20 cm high filled with water (diameter does not matter, but small is better if you want to install a “back flow safety, see below))
c) a rubber band / a clamp
(Follows from Dr. von Reuss description)
Assembly: Place the hose into the glass with water, as deep as you need pressure, eg. p= 10 cmH2O, the opening of the hose needs to be 10 cm deep. p= 20 cmH2O => the opening needs to be 20 cm deep. Fix the hose in the appropriate depth with the rubber band.
In case you want a “back flow safety” you need to install above the glass a ball (a hollow sphere), on top a connection for the hose from the mouth of the patient, at the bottom the hose goes to the water glass.
The ball needs to have a volume which is greater then the filling of the glass.
(I’m a chemist, and this is a very common apparatus in chemistry).
The setup can also be printed and has the advantage, it is from a mechanical point of view quite easy to setup, you don’t need the steal springs nor a different material for the ventile. You can print it with cheap PLA.
If you give me the diameter of the hose, I will send you a stl file.
(I’m a chemist, and this is a very common apparatus in chemistry).March 30, 2020 at 16:49 #99022
We are also following this approach, but the fact of having a 30cm high vessel with water is not something very desirable on a hospital we think. Stability an posible tilting makes us aware of the risks this design.
One possible solution could be not using water, but another fluid with more viscosity so the size of the system could be smaller and be properly confined.
What do you think on this?
ThanksApril 1, 2020 at 09:55 #99102AndrésParticipant
All, we are a group of Aerospace Engineers who are also working on a design for this valve.
If you are working on something like this, you should definitely take a look at this patent: https://patents.google.com/patent/US8534283B2/en?q=peep&q=valve&oq=peep+valve
Please keep us posted on any update you can share. I will do the same if we are able to solve it.
Andrés.April 3, 2020 at 10:05 #99209Marc JofreParticipant
Any new information on this topic PEEP valves?
Best regardsApril 4, 2020 at 23:51 #99282AndrésParticipant
We have a design ready and two prototypes already done. We are sending them to a local hospital to validate the design. Once it is ready we will make it public with all the required information for its fabrication.
We may need to calibrate the spring we are using, but we need first the feedback from the medical center.
I will keep you posted.
Andrés.April 5, 2020 at 14:22 #99293
We are also testing our prototypes, can you send me an email to email@example.com to share ideas?
Somos un grupo de ingenieros de la UPM, podemos compartir ideas.
Gracias!!April 6, 2020 at 08:37 #99300Marc JofreParticipant
Thanks for the information and work.
I’m located in Barcelona (Spain). I follow different channels to check for updates. I encourage you to make the highest diffusion possible when ready!
Hope it goes all well.
MarcApril 6, 2020 at 19:37 #99345Natan Pereira DornelesParticipant
I’m from the federal univerity of Rio Grande do Sul in Brazil (UFRGS) and we started working on a PEEP valve today. It would help a lot if there was already a validated design. Waiting for your update, Andrés!
Thank you so much!April 8, 2020 at 20:40 #100712IanParticipant
Hi, I work within a Research centre at a University and would also be interested when the valves have been validated We are working on a number of Covid 19 projects and there is definitely a need for these as supplies are low.April 10, 2020 at 12:28 #104942Ankit AgarwalParticipant
We have been working on all the elements of the design and are now into 3D printing and trial at a hospital in India. We have a good panel of doctors closely working with us to ensure we can take this to large scale production.
I would suggest have a global Zoom call to discuss our design and hear any suggestions. Can we have the call at 5PM (India Standard Time) on this Saturday- 10th April.
Please email me on firstname.lastname@example.org for confirmation. I will share the zoom details.April 11, 2020 at 06:52 #105147RedfluidParticipant
My name is Enric from company Precision valve company Redfluid. Together with several hospitals in Barcelona we have developed a high precision PEEP valve with excellent results in lab trials and health patients. We are awaiting last approval from the government. We are looking for preorders so we can get funding to mass produce the Peep valves. We have a huge production capacity so everyone should be able to get a Peep valve.
please if you know anyone interested in Peep valves kindly contact us.
- You must be logged in to reply to this topic.